Unloader One X OTS
Take control of knee joint pain, quickly and easily with the Unloader One X. When properly applied, the brace will reduce load on the painful and affected side of the knee through a patented unloading system.
1. With knee slightly bent, place the brace on your leg with the hinge on the affected side of your knee. Align the hinge so that the Ossur logo is at the top of your kneecap and the front of the hinge is at the center of the side of your leg.
2-5. Close the lower buckle (blue dot) first, followed by the top buckle (yellow dot). Make sure that the calf strap is positioned above the widest part of your calf. Use the palm of your hand to close the buckles, pressing firmly until you hear them click.
6. Your brace should be set to provide adequate pain relief with the SmartDosing® scale set at the middle indicator. However, depending on your level of pain or volume fluxuations due to weight gain/loss or swelling, you can adjust the strap tension easily by turning the dials. Turning the dials clockwise (the indicator moving toward the.+) will increase the tension while turning the dials counter clockwise (the indicator moving toward the -) will decrease the tension. Remember to keep your leg slightly bent when using the SmartDosing® dials.
7. Check the final position of the brace on your leg. When your leg is slightly bent, the Ossur logo on the hinge should be aligned with the top of your kneecap and the front edge of the hinge cap should be at the center of the side of your leg.
Hint: If you find it hard to align the brace and buckle it at the same time, simply place the brace on your slightly bent leg, put the buckles in the correct receiving keyholes WITHOUT closing the buckles, and pull up your leg into place.
Once aligned correctly, continue with step 2-7.
When starting to use the brace, it is recommended to gradually build up to full time use, so the skin can get used to the brace.
Skin irritation issues may occur for some users. If you have sensitive skin, it is recommended to use the Unloader One® X as tolerated.
- If redness appears, usage should be decreased.
- If problems persist, discontinue brace use and consult your health care professional or practitioner regarding other liner options.
If no skin issues arise you can build up to full time usage more quickly.
How to Remove the Unloader One® X:
Open both upper and lower Quick Fit Buckles.
Remove the buckles from their keyholes.
Remove the brace from your leg.